Bubble Emission Test
The Bubble Emission test determines package integrity for porous and non-porous materials. It meets the requirements of regulatory agencies to demonstrate package integrity. Nelson Laboratories complies with ASTM 2096 and ISO 11607 “Packaging for terminally sterilized medical devices,” and we are actively involved in the ASTM committees.
- ASTM F2096
- ISO 11607
Ask an expert for a specific consultation on your product. If you are ready to submit your samples for testing, click here to fill out the Sample Submission Form.
|PKG260||Packaging: Bubble emission, ASTM F2096, per pouch||Add|
PKG260 - Packaging: Bubble emission, ASTM F2096, per pouch
Study OutlineThe Bubble Emission test begins by setting the pressure gauge to 10 + 2 inches of water. A packaging port is attached to the center of the sample and a probe is inserted through the port. The package is then submersed into an immersion fluid and inflated. Evidence of bubble emission through the package is considered a failure.
Please see our Packaging Validation matrix for more information.