Antibiotic Potency Test
The Antibiotic Potency test measures the bioactivity or potency of various antibiotics. All antibiotics must go through potency testing prior to market release. Nelson Laboratories has experience testing many different antibiotics, including Vancomycin, Gentamicin and Amphotericin B.
This test is done in compliance with the cylinder plate method described in the USP General Chapter 81 on Antibiotics-Microbial Assays. Our scientifically sound testing includes tight controls that can give clients a high level of confidence in the test results.
- USP 81
Ask an expert for a specific consultation on your product. If you are ready to submit your samples for testing, click here to fill out the Sample Submission Form.
|APA110||Antibiotic Potency Assay, USP 81||Add|
|APA115||Antibiotic Potency Assay, USP 81, additional samples||Add|
APA110 - Antibiotic Potency Assay, USP 81
TAT: 35 days
APA115 - Antibiotic Potency Assay, USP 81, additional samples
Study OutlineIn the Antibiotic Potency test procedure, cultures are grown and adjusted using turbidimetric measurement techniques. An aliquot of the adjusted culture is added to a thin layer of agar to create a seed layer. Test samples are diluted to an appropriate test concentration according to labeled potency claims. A reference standard antibiotic is diluted in a similar manner with several dilutions used to create a standard curve.
For each sample tested, three plates are used. Six stainless steel penicylinders are placed on each plate so that they are at approximate 60° intervals with a center spacing of approximately 5 cm. Three cylinders are filled with the median dilution standard and three cylinders are filled with the sample. The plates are incubated and zones of inhibition are measured with calibrated calipers sensitive to 0.01 mm.
A standard curve is prepared the day of testing. For the standard curve, three plates are prepared for each of four concentrations that bracket the median test dilution. Six cylinders are placed on each plate so that they are at approximate 60° intervals with a center spacing of approximately 5 cm. Three alternate cylinders are filled with the median test dilution and the other three cylinders are filled with the dilution concentration of the standard curve. The plates are incubated and zones are measured with calibrated calipers sensitive to 0.01 mm.
The measured zone values are entered into a validated spreadsheet for zone diameter correction, linear regression output and potency calculations. Using a log transformation straight line method with a least squares fitting procedure, a reference point can be interpolated. Log-linear regression is used to compute an R2 value to estimate linearity.
The sample potency is estimated by averaging the reference standard zone diameters and the sample zone diameters on the three plates used. Concentrations are calculated from the corresponding corrected values of zone diameters. The log value is converted to the antilog. The antilog value is multiplied by the dilution factor to obtain the concentration in mg/ml of active Gentamicin. These calculations are done with a validated spreadsheet for antibiotic potency.